Perbedaan Kadar Hemoglobin Metode Sianmethemoglobin dengan dan Tanpa Sentrifugasi pada Sampel Leukositosis
Keywords:sianmethemoglobin methods, centrifugation, leukocytosis
AbstractExamination of hemoglobin levels influenced leukocytosis sianmethemoglobin method that causes increased absorbance measurements of hemoglobin levels increased significantly and the false blood sample that has been diluted with a solution Drabkins in centrifugation at 3000 rpm for 10 minutes and then the absorbance of the supernatant was measured with a photometer at λ 546 nm. This study aimed to analyze the differences in hemoglobin level examination siamethemoglobin method with and without centrifugation at sample leukocytosis. This type of research is observational research laboratory. The study design was cross-sectional study. Samples were taken from the remaining blood samples of patients who have been examined leukositnya number more than 20,000 / uL with Hematology Analyzer (CEL-DYN Ruby) February-April 2014, and were divided into 4 groups based on criteria that group 1. leukocyte count of 20,000 / uL-29 999 / mL, group II. 30,000 / uL-39 999 / uL, the group III. 40,000 / uL-49,999 / uL, the group IV. More than 50,000 / uL. The number of samples taken were 20 samples of each group, a total sample of 80 samples. The analysis showed no significant difference in hemoglobin levels siamethemoglobin method with and without centrifugation at sample leukocytosis with a value of p = 0.000 less than 0.05 α. Leukocytosis Turbidity affects the difference in hemoglobin levels with and without centrifugation, the higher the number the greater the difference in leukocyte levels of hemoglobin, hemoglobin level examination results of the study based on the criteria of the number of leukocytes obtained by the difference in hemoglobin levels with and without centrifugation in group I. 0.22 ± 0.07 g / dL, group II 0.40 ± 0.22 g / dL, a group III. 0.44 ± 0.14 g / dL, Group IV. 0.85 ± 0.41 g / dL. The level of hemoglobin in the sample sianmethemoglobin method leukocytosis with more than 20,000 / uL need a centrifuge so that appropriate hemoglobin levels over the patient's clinical condition.
How to Cite
Publishing your paper with Medical Laboratory Technology Journal (MLTJ) means that the author or authors retain the copyright in the paper. MLTJ granted an author(s) rights to put the paper onto a website, distribute it to colleagues, give it to students, use it in your thesis etc, even commercially. The author(s) can reuse the figures and tables and other information contained in their paper published by MLTJ in future papers or work without having to ask anyone for permission, provided that the figures, tables or other information that is included in the new paper or work properly references the published paper as the source of the figures, tables or other information, and the new paper or work is not direct at private monetary gain or commercial advantage.
MLTJ journal provides immediate open access to its content on the principle that making research freely available to the public supports a greater global exchange of knowledge. This journal is licensed under a Creative Commons Attribution-ShareAlike 4.0 International License. This license lets others remix, transform, and build upon the material for any purpose, even commercially. MLTJ journal Open Access articles are distributed under this Creative Commons Attribution-ShareAlike 4.0 International License (CC BY-SA). Articles can be read and shared for All purposes under the following conditions:
BY: You must give appropriate credit, provide a link to the license, and indicate if changes were made. You may do so in any reasonable manner, but not in any way that suggests the licensor endorses you or your use.SA: If you remix, transform, or build upon the material, you must distribute your contributions under the same license as the original.